Principle of Operation :
Level transmitter works on the principle of reed resistance/permanent magnet basis. Reed switches
function like the center arm (as shown in fig. 1) of a potentiometer. As per liquid level float moves up or down, magnet inside the float operates the reed switches thereby causing a change in resistance value. Change in the resistance is measured by the electronics unit and conditioned as a current output of 4 to 20 mA. Signal conditioner is head mounted and of 2-wire loop powered suitable for 12 to 24 VDC giving output of 4 to 20 mA.
Technical Specification : Our Standard Product
Model (VMT-04-F65-AW) , (VMT-04-F92-AW) top mounted magnetic reed/resistance cascade type loop powered 2 wire level transmitter. Process Connection flange 78 or 120 mm diameter with 4 bolt holes of 6mm or 14 mm dia on 65 or 92 mm PCD respectively. Wetted parts flange & float stopper in SS 304. Float material SS 316L and diameter 41, 53 or 68 mm. Operating pressure 10kg/sq.cm and Temperature 125 deg.C. Maximum.
Salient Features :